This instrument is designed and made as per national standard ASTM D1665 and GB/T266 Petroleum Products- Determination of Engler Viscosity, T0622 Bitumen Engler Viscosity Determination (Engler viscometer method) in national industrial standard JTG E20-2011 Standard Test Methods of Bitumen and Bituminous Mixtures for Highway Engineering and Shanghai enterprise standard Q/YXYY11 WNE-1 Engler Viscometer. It is suitable to determine the rate of time(seconds) of liquid flow out from Engler viscometer and time(seconds) of distilled water flow out at 20℃ under certain temperature and cubage. This rate is the Engler viscosity. Unit is Engler degree.
I. Main technical features
1. The instrument adopts desktop, all-in-one and double inner crucible structure. Both of inner and outer crucible adopt stainless materials. Inner crucible has been specially processed. The smooth finish corresponds with the test requirements.
2. The flow out time adopts digital display. Can do parallel determination for 2 samples at the same time. Easy to use and efficiency is high.
3.The instrument adopts the semiconductor refrigeration piece.
II. Main technical specifications
1. Power supply: AC(220±10%)V, 50Hz
2. Standard water value: K20= (51±1)s, (when the sample is asphalt: K20×0.224)
3. Temperature range: (20～100)℃
4. Temperature controlling: (20±0.1) ℃, (25±0.1) ℃
5. Temperature control accuracy: ±0.1℃
Add: RM. 803, Chevalier House, 45-51, Chatham Road South, Tsim Sha Tsui, Kowloon, Hong Kong